3-1/2 Digit Backlit LCD Terminal Board Pressure Transmitter; 0 to 1000 psi
Using 360° Viewing:
- Rotate: Use top-to-bottom, side-to-side by use of mouse arrow.
- Zoom In: Double click on image.
- Zoom Out/Reset: Put photo at full zoom & then double click.
- Transmitter Type Pressure Transmitter
- Wetted Materials - Transmitters 316 Stainless Steel
- Range Values - Transmitter 0 to 1000
- Range Units - Transmitter psi
- Accuracy - Transmitter +/-0.25% Full Scale
- Electrical Connections - Transmitter Terminal Board
- Power Required - Transmitters 12 to 32VDC
- Process Connection Size - Transmitter 1/4"
- Process Connection Type - Transmitter FNPT
- Shape - Transmitter Irregular
- Overall Height - Transmitter 3.35"
- Overall Depth - Transmitter 3.62"
- Diameter - Transmitter 2.56"
- Connection Location - Transmitter Bottom
- Output - Transmitter 4 to 20mA DC
- Display - Transmitter 3-1/2 Digit Backlit LCD
- Operating Temp. - Transmitter 14 Degrees to 140 Degrees F
- Wiring Entry - Transmitter 1/2" FNPT
- Features Push Button Configuration, Rotatable Display, Min/Max Hold, IP65 Rating
- Loop Resistance Dependent on Loop Supply Voltage
- Housing Aluminum
- NEMA Rating 4X
- Max. Pressure (PSI) 2000
- Thermal Error +/-0.2%
- Includes Instructions
- Manufacturers Warranty Length 1 yr.
- Display Units PSI
Compliance and Restrictions
Alternate Products may not be identical in style or function to original selection. Please compare below for more information.
Item # 5DDD3ASHCROFT
Product Reviews Disclaimer:
Grainger is neither responsible for, nor does it endorse, the content of any product review or statement posted. Any statements posted constitute the statements of the poster and are not the statements of Grainger. The statements posted by Grainger employees with the Grainger employee badge represent the views of such employees and are not the statements of Grainger. Grainger makes no representations as to the appropriateness, accuracy, completeness, correctness, currentness, suitability, or validity of any product review or statements posted, including those posted by employees with the Grainger employee badge, and is not liable for any losses, injuries or damages which may result from any such product review or statements. Use of any linked web site provided in a product review or post is at the user's own risk.