1/4" MNPT Explosion Proof Pressure Transmitter 0 to 15 psi, 4 to 20mA DC Output
Using 360° Viewing:
- Rotate: Use top-to-bottom, side-to-side by use of mouse arrow.
- Zoom In: Double click on image.
- Zoom Out/Reset: Put photo at full zoom & then double click.
- Transmitter Type Explosion Proof Pressure Transmitter
- Wetted Materials - Transmitters 316L Stainless Steel
- Range Values - Transmitter 0 to 15
- Range Units - Transmitter psi
- Accuracy - Transmitter +/-0.25%
- Electrical Connections - Transmitter 3 ft. Cable
- Power Required - Transmitters 12 to 30VDC
- Process Connection Size - Transmitter 1/4"
- Process Connection Type - Transmitter MNPT
- Output - Transmitter 4 to 20mA DC
- Operating Temp. - Transmitter -40 Degrees to 257 Degrees F
- Safety Rating - Transmitter Class I, Div. 1 and 2, Groups A, B, C, D; Class I and II, Div 1 and 2, Groups E, F, G
- Wiring Entry - Transmitter 1/2" FNPT
- Housing 304 Stainless Steel
- NEMA Rating 7 and 9
- Loop Resistance Dependent on Loop Supply Voltage
- Max. Pressure (PSI) 30
- Thermal Error +/-2% (-4 to 185F)
- Compatible With Any Process Compatible with 316Stainless Steel
- Includes Instructions
- Manufacturers Warranty Length 1 Year
Compliance and Restrictions
Item # 5DDP6ASHCROFT
Grainger is neither responsible for, nor does it endorse, the content of any product review or statement posted. Any statements posted constitute the statements of the poster and are not the statements of Grainger. The statements posted by Grainger employees with the Grainger employee badge represent the views of such employees and are not the statements of Grainger. Grainger makes no representations as to the appropriateness, accuracy, completeness, correctness, currentness, suitability, or validity of any product review or statements posted, including those posted by employees with the Grainger employee badge, and is not liable for any losses, injuries or damages which may result from any such product review or statements. Use of any linked web site provided in a product review or post is at the user's own risk.